Here you will find the answers to frequently asked questions about Elos Accurate® including Elos Accurate® Hybrid Base Bridge, Elos Accurate® Hybrid Base Single, and Elos Accurate® Scanbody as well as Elos Accurate Model Analog.
No, not yet. At the moment you can only use the Elos Accurate Prosthetic Screwdriver for Elos Accurate® Hybrid Base Bridge.
You should use the screwdriver from the implant manufacturer.
Soon. We are in the phase of developing an Elos Accurate® Hybrid Base Single for angled screw channels.
If you have bought an Elos Medtech dental product from one of our distributors, please contact your local salesperson for more information. The distributors usually offer a warranty for implant losses. In general, the Elos Medtech Guarantee covers a 10-year warranty for all of our Elos Accurate® and Elos Cresco® products.
Our Elos Accurate® Hybrid Base Bridge is an original concept, invented by Elos Medtech, thanks to its unique features and advantages.
Our Elos Accurate® Hybrid Base Single is an original concept, invented by Elos Medtech, thanks to its unique features and advantages.
If one or both of the antennas break, you will lose the grip function. This is however not important for the long-term result of the prosthetic restoration; you can still use Elos Accurate® Hybrid Base without risk! If the antennas bend, you may bend them back to their original position.
If the Elos Accurate® Prosthetic Screw breaks in the patient’s jaw, try first to use an ultrasonic sound to take away the screw remains. If not successful, we recommend you to refer the patient to a specialist clinic.
No, you don’t. You can use your Talladium screwdriver. However, we have made some improvements to our screwdriver, which is now intended for up to 35 Ncm, allowing you to feel more secure when you torque to required Ncm set by the implant companies.
Yes, we do! Elos Medtech has more than 40 years of experience within the dental industry. We have an original concept with an implant interface compatible with the major implant systems. The special features of Elos Accurate® are based on our vast experience and in-depth knowledge in developing and manufacturing dental products. The features of Elos Accurate® Hybrid Base are founded on our passion for innovation. Our design philosophy is to simplify digital dentistry with full flexibility in an Open Digital Workflow!
It means that the maximum angle of the screw channel is 28° and that the dental implants can be tilted up to 30°.
For Elos Accurate® Hybrid Base, we have created a new library that is free to download from our website. It is important that you use our library structure instead of scanning the physical component. We have all the needed parameters and features, for example, the pre-set cement gap, included in the library.
We work with the three largest software 3Shape, Dental Wings and exocad.
For Sirona users: You are not able to install our libraries in the Sirona software.
Blast the inner geometry of the milled dental bridge interfacing the Elos Accurate® Hybrid Base with aluminium oxide ≤ 50 µm and blasting pressure of two bars. Before cementing, clean the surface thoroughly with alcohol. Note: Blasting of the Elos Accurate® Hybrid Base Hybrid Base™ Bridge is not necessary. For more information, download our IFU.
We recommend you to use 3M ESPE RelyX™ self-adhesive cement since this is the cement we have used throughout our process when developing and testing our product.
The Elos Accurate Scanbody can be used both for taking a digital impression using an intra-oral scanner, and for taking a digital impression using a lab scanner. We are currently in the phase of removing our desktop scanbody.
You probably use too much force when tightening the screw. You should always tighten lightly by hand and no more than 5 Ncm.
The product lifetime is 100 scanning cycles or maximum two years.
You can use the same model analog for printed, milled or conventional plaster models.